Shanghai Haozhuang Instrument Co., Ltd. develops the LNB brand biochemical incubator, which is energy-saving and environmentally friendly, leading the country. Professional manufacturing is more secure and affordable
Instructions for use: 1. Place the incubator on a flat and solid ground, adjust the two supporting screws at the lower end of the box to make the box stable. 2. Plug in the power socket (the power supply should be well grounded), press the "power switch", the display is on, at this time the display shows the actual temperature inside the incubator and its working time. 3. Time setting: Time setting includes the setting of "minute" and "hour". Press the "SET" setting key, when the decimal point in the lower right corner of the display position of the "minute" digital tube lights up, that is, enter the "minute" setting state, and then press the "â–²" or "â–¼" key to confirm the "incubator" work this time. Minutes â€time (the longest is 59 minutes); press theâ€œ SET â€key again, when the decimal point in the lower right corner of theâ€œ hour â€nixie tube display is lit, it enters theâ€œ hour â€setting state, then pressâ€œ â–² â€orâ€œ â–¼ "Key to confirm the" hour "time (maximum 99 hours) of the incubator's current work. 4. Temperature setting: Press the "SET" key, when the decimal point in the lower right corner of the last digit of the temperature display lights up, it enters the temperature setting state, and then press the "â–²" or "â–¼" key to confirm the incubator. Sub-set temperature (set temperature range is 5 â„ƒ ï½ž 50 â„ƒ). When the above steps 3 and 4 are completed, press "ENTER" to confirm the working time of the incubator and the working temperature (set temperature) in the incubator. Note: After the temperature setting is confirmed, the temperature cannot be set back and forth at random, so as to avoid the compressor starting frequently, causing the compressor to overload, and affecting the service life of the compressor. 5. If you need to check the working time and temperature set by the incubator this time, press the "SET" key, the display panel will display the set time and temperature, and then press the "ENTER" key, the display value of the incubator returns To the original working state. 6. If lighting is needed in the incubator, press the "lighting switch"; if lighting is not needed in the box, the lighting switch on the panel should be placed in the "off" position to avoid affecting the upper temperature.
Matters needing attention and maintenance: 1. When the use temperature is low, condensate will be generated in the incubator, and the accumulated water in the bottom water tray at the bottom of the box should be drained regularly. 2. In order to maintain the beauty of the equipment, do not use acid or alkali and other corrosive materials to wipe the surface of the box, and dry the inside of the box with a dry cloth regularly. 3. The power supply used must have a reliable ground wire to ensure that the ground wire of the incubator is in reliable contact with the ground wire of the network power supply to prevent damage caused by electrical leakage or accidental network power supply. When the incubator is stopped, the power plug should be unplugged. 4. The minimum distance between the incubator and the wall should be greater than 10cm to ensure good cooling of the refrigeration system. 5. The room should be dry, well ventilated, the relative humidity should be kept below 85%, there should be no corrosive substances, avoid direct sunlight on the incubator
Shanghai Haozhuang Instrument Co., Ltd. is a variety of biochemical, light incubator, shaking incubator, constant temperature incubator, mold incubator, hot and cold cycler, shaker, oscillator, stirrer, water bath, centrifuge, electric heating plate. Professional manufacturing enterprise. The company integrates the research, development, production and sales of high-tech experimental instruments. Its products have eight series and more than one hundred varieties. Relying on modern science and technology, the company continuously improves product quality and launches new products to keep it up to the international level. The products are exported to all parts of the country, and have won the trust of users with excellent quality and perfect service. The employees of the company have always pursued the enterprise spirit of "unity, truth-seeking and innovation". With the tenet of "user first, quality first, reputation first", we sincerely welcome new and old customers to visit us.
Tea Pouch Packing Machine,Multi-Function Bag Making Machine,Multifunctional Bag Making Machine,Multifunctional Bag Making Machine
Wuxi Zhengyu Technology Development Co.,Ltd. , https://www.zhengyumachine.com